LabVIEW could read back some number of bytes but still throw a timeout error. Thanks for your reply! If the value is zero, than no voltage can “flow” and your supply goes into limiting mode. Hull Law Block Diagram. Message 5 of 5.
|Date Added:||16 April 2009|
|File Size:||43.23 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Agilent Technologies / Keysight Technologies E3631A
You used the word “connected”, but did not say if you where sending and receiving messages. What kind of USB labviww serial device are you using?
That really was the mistake. Message 3 of 8. Auto-suggest helps you quickly narrow down your search results by suggesting possible matches as you type. The BK was working all along with its own adapter. After that I had to remove the error queries from the Configure Output and the Configure Current Limit as well, which made these work too.
Message 7 of 8. We fix that by having the software go out and hunt for the instrument on what ever comport it is on.
I fear this might not be fast enough for my system. I do not have the power supply, labvirw my guess is that this boolean when set to true, only allows the predefined values.
Agilent E363X Series Power Supply
You can’t tell if you are “connected” to a device until you actual try to send and receive messages. The remote interface can be selected from the front-panel only.
I somehow think it would be a bad idea to just disable all the stuffs and hope fopr the best I did not know this was not accepted and did not plan on playing someone off against someone else. All you code contains links to the drivers installed so anyone without the drivers installed cannot help you by looking at you code.
Labview -> HP EA PowerSupply Problems – Discussion Forums – National Instruments
Message 8 of E36631a, just substitute your drivers if different. Set and Read the voltage. This seemed to work, but the next error appeared when it tried to reset.
Can you tell me what the problem is?? I have tried to copy your VI Omar, but somehow I couldn’t get it to run, I really don’t know why, but everything I get is the error. Will try to communicate with a null modem in between.
More than likely what’s happening is that the code is jumping between a read and write while it’s in the middle of reading. Message Edited labvoew James W on Or do you think I should get myself another adapter?
Solved: Agilent EA – Page 2 – Discussion Forums – National Instruments
Message 12 of Measuring Current with Agilent EA. Auto-suggest helps you quickly narrow down your search results by suggesting possible matches as you type. Message 5 of Thank you very much for the example for setting the default values myself.
The first cable I tried did not work. Message 3 of Try looking at the manual on the agilent website http: E363a did not know that and it helped a lot! Message 10 ph Message 5 of